The figure is below.

The equilibrium condition for the bridge is `V_(CD) = 0 "Volts"` .

On the branch ACB of the bridge we have

`I_(ACB) = U/(R1+X_(C1) +R3)`

`V_(AC) = I_ACB*R1 = U* (R1)/(R1 +X_(C1) + R3)`

On the branch ADB of the bridge we have

`I_(ADB) = U/(R2 +X_(Cu))`

`V_(AD) =I(ADB)*R2 = U*(R2)/(R2+X_(Cu))`

As said above for equilibrium `V_(CD) = 0 "Volts"` which means

`V_(AC) = V_(AD)`

`(R1)/(R1+X_(C1) +R3) = (R2)/(R2+X_(Cu))`

or equivalent

`(R1)/(R3+ X_(C1)) = (R2)/X_(Cu)`

The above equation leads to

`(R1)/(j*omega*Cu) - (R2)/(j*omega*C1) = R2*R3`

Since the lest term is pure imaginary and right term is real, first condition that needs to be met for bridge equilibrium is R3=0.

Now we can write the second condition for bridge equilibrium as

`(R1)/(Cu) = (R2)/(C1)`

`Cu = C1* (R2)/(R1) =0.5*10000/3200 =1.5625 microF`

Observation:

In practice the resistance R3 in series with C1 is provided to compensate for resistive looses in the unknown capacitor Cu. Therefore to measure Cu, first we modify the value of R3 for minimum indication of Voltmeter V, then we modify the value of R1 until the indication of voltmeter is zero.

**Further Reading**

## We’ll help your grades soar

Start your 48-hour free trial and unlock all the summaries, Q&A, and analyses you need to get better grades now.

- 30,000+ book summaries
- 20% study tools discount
- Ad-free content
- PDF downloads
- 300,000+ answers
- 5-star customer support

Already a member? Log in here.

Are you a teacher? Sign up now